Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
|Published (Last):||13 April 2015|
|PDF File Size:||7.36 Mb|
|ePub File Size:||19.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
The remaining free space in the media is displayed at the lower right of the screen.
Manual Cardiofax 1150
The file may be damaged. Action Enter correct values. In this test, the input analog signal processing circuit lehislativo 1 mV rectangular waveforms that are printed. You cannot charge the battery when the power is turned on.
Otherwise the internal temperature of the electrocardiograph rises, which leads to inaccurate operation and shortens the electrocardiograph life. At 80 MHz and MHz, the higher frequency range applies.
This warranty does not apply to products that have been modified, disassembled, reinstalled or repaired without Nihon Kohden approval or which have been subjected to neglect or accident, damage due to accident, fire, lightning, vandalism, water or other casualty, improper installation or application, or on which the original identification marks have been removed.
Otherwise, the SD memory card may become unusable. Clean the gray colored part of the thermal head with the thermal head cleaner pen.
Apply a thin layer of cardioCream on the electrode surface. Press the Cancel function key to cancel. The name of the pressed key is displayed on the screen. Press the function key. There is not enough space to save new ECG data file in the memory. If the thermal head gets contaminated, clean it with the thermal head cleaner pen.
Thermal head 30 km continuous recording When the thermal head wears out, the recording density gets weaker due to missing dots. Press the or function key to select the desired item. There are two types of data save format, type 1 normal and type 2 compressed.
Select the number of times to acquire ECG waveform. Push the paper magazine release button and open the paper magazine. Turn the power off after processing is complete.
decreto legislativo pdf – PDF Files
This is used to check if the input analog signal processing circuit detects the lead off condition correctly using the input check jig. The hanger prevents the cables from getting tangled or the limb leads from getting in the way. Set the recording paper in the recorder cover by spreading the paper guides. Only use new electrodes.
We recommend saving and printing the settings immediately after installation. Select another media and save the files again. Attach the battery cover and secure it with the screw. Legixlativo disinfect the electrode, wipe it with a soft cloth moistened with any of the disinfectants listed below. If another cart is used, it may tip over or the electrocardiograph may fall off.
If it is almost discharged, operate the electrocardiograph on AC power or recharge the battery. The electrocardiograph is shipped from the factory with no battery pack installed. Recorder cover release button 2. Changes the character type.
Electrodes with a deteriorated surface may cause high contact impedance and distorted ECG waveforms. The LEDs on the leguslativo panel light up one by one. The ECG analysis result is incorrect if artifact is superimposed dexreto the notselected leads or interval. Select the age group by pressing the AGE key.
Error Message Battery trouble Cause The battery contact voltage is high. Nothing can be printed. Turning the Power On Before turning the power on, check the following items. Press any key to turn the screen on again. Otherwise, the card might not operate. Before disposing of the battery pack, check with your local solid waste officials for details in your area for recycling options or proper disposal. Change the search condition and search again.
decreto legislativo 1150 pdf
To cancel printing, press the Cancel function key. Failure to follow this warning may cause electrical shock or injury to the patient. Press the Start function key. If you frequently charge the battery after it has only been used for a short time, it may shorten the life of the battery.